Read Online or Download ANSI IEEE C63.5 ( [Electromagnetic Compatibility-Radiated Emission Measurements in Electromagnetic Interference -EMI- Control-Calibration of Antenas. 9 kHz to 40 GHz] [EMC] [73pages] PDF
Similar measurements books
The main accomplished and up to date reference paintings masking the clinical basics and useful issues for designing, constructing and enforcing measuring platforms in a wide diversity of engineering and technological fields. Measuring structures lie on the middle of all technological know-how, engineering and expertise, and are used to quantify actual phenomena corresponding to temperature, strain, voltage and present.
Foundations of size bargains the main coherently equipped remedy of the themes and matters vital to dimension. a lot of the study concerned has been scattered over a number of many years and a mess of journals--available commonly purely to specialties. With the booklet of Volumes and 3 of this significant paintings, Foundations of size is the main finished presentation within the zone of size
"The fact belongs to every body, yet mistakes is ours on my own. " -- The degree of All issues Amidst the chaos of the French Revolution, intrepid astronomers set out in contrary instructions from Paris to degree the area, one voyaging north to Dunkirk, the opposite south to Barcelona. Their findings could aid outline the meter as one ten-millionth of the space among the pole and the equator, a typical that has on account that swept the planet.
- Particle Size Measurements: Fundamentals, Practice, Quality
- High-Voltage Test and Measuring Techniques
- Computational Methods and Experimental Measurements XV (Wit Transactions on Modelling and Simulation)
- The Hawker Siddeley Trident
Extra resources for ANSI IEEE C63.5 ( [Electromagnetic Compatibility-Radiated Emission Measurements in Electromagnetic Interference -EMI- Control-Calibration of Antenas. 9 kHz to 40 GHz] [EMC] [73pages]
The contribution from the source, resulting from the analyzer and signal source not being tuned to the same frequency, can be computed as the sum of the signal source frequency accuracy and the spectrum analyzer frequency accuracy times the slope of the antenna factor at the frequency of interest. For some antennas the use of a spectrum analyzer, automated tuned receivers and spectrum analyzer/receiver hybrids may require more than one value of uncertainty to be stated as a function of frequency due to the antenna factor slope.
The leads shall be kept as short as possible and kept close to the surface of the metal box or frame. A spacing of 5 mm to 10 mm is suggested. 1 shows an example. The T-connector used in the antenna factor measurement setup may be built into the dummy antenna box. Also, a resistor network to provide matching to the generator and the RF voltmeter (receiver) may be built into the dummy antenna box. 30 Copyright © 2006 IEEE. All rights reserved. , BNC Connector to mate with rod input port C S C: 10 pF 5 %, Silver Mica.
These sources are amplitude accuracy and frequency accuracy. An antenna calibration may be conducted in either a swept method or discrete frequency method. For both methods, the amplitude uncertainty contribution (uASpA) is based on the instruments relative amplitude uncertainty as a function of the reference level. This 54 Copyright © 2006 IEEE. All rights reserved. 5-2006 RADIATED EMISSION MEASUREMENTS IN EMI CONTROL—CALIBRATION OF ANTENNAS (9kHz TO 40 GHz) contribution may be assessed using either the instruments specifications or through a type A evaluation.